The Oregon Manufacturing Innovation Center Research and Development (OMIC R&D) recently welcomed Capture 3D, a Zeiss company, as a member of the collaboration, now at 41 members. Headquartered in Santa Ana, California, Capture 3D is a leader in innovative optical 3D metrology solutions, including accurate blue light 3D scanners, precision photogrammetry systems, comprehensive inspection software, and high-tech automated inspection systems.
In this article, equations for finding profile and base pitch errors with a micrometer are derived. Limitations of micrometers with disc anvils are described. The design of a micrometer with suitable anvils is outlined.
Base helix error - the resultant of lead and profile errors is the measured deviation from the theoretical line of contact (Fig. 1). It can be measured in the same way that lead error on a spur gear is measured, namely, by setting a height gage to height H based on the radial distance r to a specified line of contact (Fig. 2), rotating the gear so as to bring a tooth into contact with the indicator on the height gage, and then moving the height gage along two or more normals to the plane of action. The theoretical line of contact on helical gear must be parallel to the surface plate, which is attained by mounting the gear on a sine bar (Fig. 3).
Our research group has been engaged in the study of gear noise for some nine years and has succeeded in cutting the noise from an average level to some 81-83 dB to 76-78 dB by both experimental and theoretical research. Experimental research centered on the investigation into the relation between the gear error and noise. Theoretical research centered on the geometry and kinematics of the meshing process of gears with geometric error. A phenomenon called "out-of-bound meshing of gears" was discovered and mathematically proven, and an in-depth analysis of the change-over process from the meshing of one pair of teeth to the next is followed, which leads to the conclusion we are using to solve the gear noise problem. The authors also suggest some optimized profiles to ensure silent transmission, and a new definition of profile error is suggested.
Much information has been written on gear inspection, analytical. functional. semiautomatic and automatic. In most
cases, the charts, (if you are lucky enough to have recording
equipment) have been explained.